INTERNATIONAL. STANDARD. IEC. First edition. Electricity metering equipment (a.c.) –. General requirements, tests and test conditions –. IEC ELECTRICITY METERING EQUIPMENT (A.C.) – GENERAL REQUIREMENTS, TESTS AND TEST CONDITIONS – PART METERING. Buy IEC Ed. Electricity metering equipment (AC) – General requirements, tests and test conditions Part Metering equipment from SAI Global.
|Published (Last):||14 March 2012|
|PDF File Size:||6.85 Mb|
|ePub File Size:||19.91 Mb|
|Price:||Free* [*Free Regsitration Required]|
Documents Flashcards Grammar checker. Metering equipment This English-language version is derived from the original bilingual publication by leaving out all French-language pages. Missing page numbers correspond to the Frenchlanguage pages. Reference number IEC Consolidated editions The IEC is now publishing consolidated versions of its publications.
For example, edition numbers 1. Information relating to this publication, including its validity, is available in the IEC Catalogue of publications see below in addition to new editions, amendments 60252 corrigenda. Information on the subjects under consideration and work in progress undertaken by the technical committee which has prepared this publication, as well as the list of publications issued, is also available from the following: On-line information is also available on recently issued publications, withdrawn and replaced publications, as well as corrigenda.
Please contact the Customer Service Centre see below for further information. The object of the IEC is to promote international co-operation on all questions concerning standardization in the electrical iiec electronic fields. Their preparation is entrusted to technical committees; any IEC National Committee interested in the subject dealt with may participate in this preparatory work.
IEC | IEC Webstore | smart grid, smart meter, smart energy
International, governmental and non-governmental organizations liaising with the IEC also participate in this preparation. The IEC collaborates closely with the International Organization for Standardization ISO in accordance with conditions determined by agreement between the two organizations.
Any divergence between the Iwc Standard and the corresponding national or regional standard shall be clearly indicated in the latter. The IEC shall not be held responsible for identifying any or all such patent rights.
INTERNATIONAL STANDARD IEC
Equipment for electrical energy measurement and load control. The text of this standard is based on the following documents: The committee has decided that the contents of this publication will remain unchanged until Electromechanical meters for active energy classes 0,5, 1 and 2 Replaces particular requirements of IEC Static meters for active energy classes 1 and 2 Replaces particular requirements of IEC Static meters for reactive energy classes 2 and 3 Replaces particular requirements of IEC Pulse output devices for electromechanical and electronic meters two wires only IEC Power consumption and voltage requirements IEC General concepts IEC Collection of meter dependability data from the field This part is a standard for type testing electricity meters.
This 26052 is intended to be used in conjunction with the appropriate part of IEC for the type of equipment under consideration.
Iev standard distinguishes between — meters intended to be used indoors and outdoors; and — protective class I and protective class II meters. The test levels are regarded as minimum values to guarantee the proper functioning of the meter under normal working conditions.
For special application, other test levels might be necessary and should be agreed upon between the user and the manufacturer. Ied equipment 1 Scope This part of IEC covers type tests for electricity metering equipment for indoor and outdoor application and applies to newly manufactured equipment designed to measure the electrical energy on 50 Hz or 60 Hz networks, with a voltage up to V.
It applies to electromechanical or static meters for indoor and outdoor application consisting of a measuring element and register s enclosed together in a meter case. It also applies to operation indicator s and test output s. If the meter has 6202 measuring element for more than one type of energy multi-energy metersor when other functional elements, such as maximum demand indicators, electronic tariff registers, time switches, ripple control receivers, data communication interfaces, etc.
It does not apply to: For rack-mounted meters, the mechanical properties are not covered in this standard. For dated references, only 6052 edition cited 620552. For undated references, the latest edition of the referenced document including any amendments applies. Current transformers IEC Inductive voltage transformers IEC 662052 terms relating to measurements — Part General terms relating to electrical measurements — Part Types of electrical measuring instruments — Part Specific terms according to the type of instrument IEC Lec — Tests A: Tests — Tests B: Dry heat Amendment 1: Tests — Test Sa: Simulated solar radiation at ground level IEC Tests — Test Fc: Vibration sinusoidal IEC Tests — Test Iev Salt mist IEC Tests — Test Ea and guidance: Tests — Test Db and guidance: Tests — Test Eh: Hammer tests IEC Symbols originals IEC Classification of groups of environmental parameters and their severities — Section 3: Stationary use at weatherprotected locations Amendment 1: Testing and measurement techniques — Section 2: Electrostatic discharge immunity test.
Testing and measurement techniques — Section 4: Testing and measurement techniques — Section 5: Surge immunity test IEC Testing and measurement techniques — Section 6: Immunity to conducted disturbances, induced by radio-frequency fields IEC Testing and measurement techniques — Section Oscillatory waves immunity test. Plastic and ebonite 3 Terms and 620522 For the purposes of this International Standard, the following definitions apply.
Expression of the performance of electrical and electronic measuring equipment has been taken from IEC Where there is a difference between the definitions in the glossary and those contained in product standards produced by TC 13, then the latter shall take precedence in applications of the relevant standard.
NOTE Standards for reactive power apply for sinusoidal currents and ifc containing the fundamental frequency only. The type may have several values of reference current and reference voltage. Each type has one designation only.
INTERNATIONAL STANDARD IEC 62052-11
NOTE 1 The type is represented by the sample meter s intended for the type tests, whose characteristics reference current and reference voltage are chosen from the values given in the tables proposed by the manufacturer. NOTE 2 Where the number of ampere-turns would lead to a number of turns other than a whole number, the product of the number of turns of the windings by the value of the basic current may differ from that of the sample meter s representative of the type.
It is advisable to choose the next number immediately above or below in order to eic whole numbers of turns.
NOTE 3 The ratio of the highest to the lowest basic speed of the rotors of each of the meters of the same type shall not exceed 1,5. Meters are designated by the manufacturer by one or more groups of letters or numbers, or a combination of letters and numbers. NOTE The type is represented by the sample meter s intended for the type tests, whose characteristics reference current and reference voltage are chosen from the values given in the tables proposed by the manufacturer.
It is usually designed and operated to obtain the highest accuracy and stability in a controlled laboratory environment 3. A single electronic display may be used with multiple electronic memories to form multiple electronic registers. For a flush-mounted meter, the meter base may include the sides of the case. It may be a single-position socket for one meter or a multipleposition socket for two or more meters 3.
It may comprise several layers which 6252 be tested singly as supplementary or basic insulation. The distortion factor is usually expressed as a percentage 3. For this range, relaxed accuracy requirements may be specified 3. It generally comprises electromagnets with their control devices.
It comprises one or more magnets and their adjusting devices. When the meter is operated from a current transformer s620552 is drawn to the need to match the current range of the meter in relation to that of the secondary of the current transformer s. The maximum current of the meter is 1,2 I n1,5 I n or 2 I n. All parts which are subject to corrosion under normal uec conditions shall be protected effectively. Any protective coating shall not be liable to damage by ordinary handling nor damage due to exposure to air, under normal working conditions.
Outdoor meters shall withstand solar radiation. NOTE For meters for special use in corrosive atmospheres, additional requirements shall be 6205 in the purchase contract for example salt mist test according to IEC The cover shall not be removable without the use of a tool.
The case shall be so constructed and arranged that any non-permanent deformation cannot prevent the satisfactory operation of the meter. The mechanical strength of the meter case shall be tested with the following tests: